<div>
	<strong>Weight and Dimensions</strong></div>
<div>
	H 13.6 cm /  Diam. 11 cm / Weight 3.0 Kg</div>
<div>
	<div>
		<strong>Transducer technical characteristics:</strong></div>
	<div>
		- Bandwidth (f-3 db) : Pressure output : 0.01-28 HZ / Pressure derivative output : DC-28 Hz</div>
	<div>
		- BLDR [0.02;4] Hz 117 dB  @ f < 1.6 Hz / 109 db @ f=4 Hz</div>
	<div>
		-Self noise 0.13 mPa/sqrt(Hz) @ 1 Hz ; < 10 dB under the LNM</div>
	<div>
		- Resolution [0.02;4] Hz 1.75 mPa<sub>RMS</sub></div>
	<div>
		- Default sensitivity: Pressure output: 20 mV/Pa /  Pressure derivative output : 2 mV/Pa.s<sup>-1</sup></div>
</div>
<div>
	<strong>Power requirments</strong></div>
<div>
	• 12 V Dc (7-20V)</div>
<div>
	• consumption : 100 mW</div>
<div>
	<strong>ENVIRONNEMENT</strong></div>
<div>
	• operating temperature : -20°c to + 50 °c</div>
<div>
	• Humidity : 100% (ip-67 with acoustic inlets sealed)</div>
<div>
	• temperature sensor : [-40°c ; +110°c]</div>
<div>
	• Absolute pressure sensor [150 ; 1150] hpa</div>