<div>
	<strong>Weight and Dimensions:</strong></div>
<div>
	H 25 cm / Diam. 11 cm / Weight 3.4 Kg</div>
<div>
	<strong>Transducer technical characteristics:</strong></div>
<div>
	- Bandwidth (f-3 db) : Pressure output : 0.01-28 HZ / Pressure</div>
<div>
	derivative output : DC-28 Hz</div>
<div>
	- BLDR [0.02;4] Hz 117 dB @ f&lt;1.6 Hz / 109 db @ f=4 Hz</div>
<div>
	-Self noise 0.13 mPa/&radic;(Hz) @ 1 Hz ; &lt; 10 dB under the LNM</div>
<div>
	- Resolution [0.02;4] Hz 1.75 mPaRMS</div>
<div>
	- Default sensitivity: Pressure output: 20 mV/Pa / Pressure derivative output : 2 mV/Pa.s<sup>-1</sup></div>
<div>
	<strong>Digitizer technical characteristics:</strong></div>
<div>
	- Clipping Level &amp; output range: Pressure: &plusmn;min (12000</div>
<div>
	[Pa/s]/2.pi.f[Hz] ; 1000[Pa]) / Pressure derivative: &plusmn;10000 (Pa/s)</div>
<div>
	- Sampling rate: 20, 50,100 Hz</div>
<div>
	- Nominal sensitivity: 1,178 10<sup>-4</sup> Pa/lsb or 1,178 10<sup>-3</sup> Pa/s/lsb @ gain= 1</div>
<div>
	- Built in gain: 1, 2, 4, 8</div>
<div>
	- Data storage: 1 GB / miniSEED</div>
<div>
	- Power requirements: 12 V DC (7-20V) / 70 mA @ 12 V</div>